Miniaturised immobilised enzymatic reactors can be used for small scale digestion of proteins. There is need for such devices; small scale devices are used either for processing of analytical sample quantities, or as proof of concept before protein digestion at larger scale. This application note compares the performance of a flow through miniaturised immobilised enzymatic reactor (μIMER) with in-solution batch digestion of simple proteins and complex matrices. Automation of peptide analysis by coupled LC-MS is explored as an option to increase throughput. In the cases evaluated, the miniaturised immobilised enzymatic reactor offered comparative results to overnight in-solution digestion, within less than 10 minutes.
Pre-activated CIMmic™ monolithic columns with 100 μL bed volume were immobilised with trypsin from bovine pancreas. This small format allows coupling to HPLC for on-line protein digestion, as well as syringe (manual) operation of the IMER. Pre-treated samples (denatured, alkylated and ultra-filtered) are injected into the column, and the eluate (tryptic digests) are subjected to LC-ESI-MS-MS analysis for protein identification and post-translational modification (PTM) determination.
Coupling trypsin enzyme onto chromatographic supports provides a platform to reuse the enzyme and automate the hydrolysis process. A monolithic chromatographic support, such as Convective Interaction Media (CIM®), enables mass transfer of molecules within its channels exclusively by convective flow. This results in enzymatic conversion which is not limited by diffusion, making CIM® monoliths ideal for the preparation of immobilised monolith enzymatic reactors (IMERs). BIA Separations offers CIMac™ trypsin IMER with a bed volume of 0.1 mL as analytical platform for mass spectrometry (MS)-based proteomics. Larger volume IMERs (up to 80 mL) are available for industrial scale production of protein hydrolysates. The following example describes the enzymatic production of β-Lactoglobulin (β-Lg) hydrolysates using monoliths with 2 μm (N1) or 6 μm (N2) average channel diameter.